ATTEN ADSCAL vs Rigol DSe were the two options left for considering . Manual, tracking, automatic cursor measurement function;. I have recently bought an ATTEN ADSCAL+ oscilloscope. During to instructions, I get the message: “Firmware update failed, error 0xFE. Buy Now Only: US$, Buy ATTEN ADSCAL Digital Oscilloscope MHz at the ADS Series User Manual Size: ( KB).
|Published (Last):||2 April 2014|
|PDF File Size:||20.9 Mb|
|ePub File Size:||3.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 79 Picture Auto mode Table Auto mode function menu: Contents in this Manual are not allowed to copy, extract and translate before being allowed by Atten Company. I use a ‘scope all the time.
Atten ADSCAL Oscilloscope Product Reviews
zds1102cal If you suspect there is damage to this product, have it inspected by qualified service personnel. The oscilloscope has seen a trigger and is acquiring the posttrigger data. Picture Save waveforms to USB flash drive, follow these steps: Probe Attenuation Setting 1.
Power cord for your country. Table Functional Menu of the Utility System 3: There is no trigger or horizontal position control of waveforms during scan mode.
ATTEN ADSCAL Digital Oscilloscope MHz
Analyze The Signal Detail 3. Save Accomplish the storage. Set trigger type of the vertical channel signal to Mode Pulse Pulse trigger. In these cases, you should use the Peak Detect mode to acquire manusl.
ATTEN ADS1102CAL Digital Oscilloscope 100MHz
The trigger signal comes from two vertical channels when you use alternative trigger. Press the top second option button.
You can select different trigger types for two vertical signals, and selected types cover edge, pulse, video and slope trigger. After the inspections above restart the oscilloscope.
ads1102xal Users can modify the name When you recall the setup, the oscilloscope will be in the mode from which the setup was saved. Edge Trigger Use Edge triggering to trigger on the edge of the oscilloscope input signal at the trigger threshold.
Atten ADS1000 Series Manual
Positive pulse width larger than pulse width setting Positive pulse width equal to pulse width setting Negative pulse width less than pulse width zds1102cal Page 54 Picture Operate Instruction: These simple examples highlight the features of the oscilloscope and give you ideas for using it to solve your own test problems. Off, 1sec, 2sec, 5sec, Infinite. Don’t have an account?
Acquire a single waveform and then stops. Select how to compare the trigger pulse relative to the value selected in the Set Pulse Width When option. You suspect that the signal contains much more detail than you can now see in the display.
Picture Operate Introduction: Software version List the software version. Picture Auto mode Table Auto mode function menu: Record Press this button to enter the waveform record menu. Support USB flash drive storage. Channel waveform and it’s Aten waveform ads1102cxl on split screen. MATH function control button. You have no items in your shopping cart.
Analyze the Humorous wave in ads1102acl Power cable.